New Automatized Method of 3D Multiculture Viability Analysis Based on Confocal Imagery: Application to Islets and Mesenchymal Stem Cells Co-Encapsulation
- 1Grenoble Alps University, Laboratory of Fundamental and Applied Bioenergetics (LBFA), Environmental and System Biology (BEeSy), Grenoble, France
- 2INSERM U1055, Grenoble, France
- 3Laboratory «Adaptations au Climat Tropical, Exercice et Santé» (ACTES; EA 3596), French West Indies University, Pointe-à-Pitre, Guadeloupe, France
- 4CNRS, TIMC-IMAG, University Grenoble Alpes, Grenoble, France
- 5Microsyst. for Biol. & Health Department, CEA-LETI, Grenoble, France
- 6Cell Therapy and Engineering Unit, EFS Auvergne Rhône Alpes, Saint Ismier, France
- 7Grenoble University Hospital, Grenoble, France
Co-encapsulation of pancreatic islets with mesenchymal stem cells in a three-dimensional biomaterial’s structure is a promising technique to improve transplantation efficacy and to decrease immunosuppressant therapy. Currently, evaluation of graft quality after co-encapsulation is only based on insulin secretion. Viability measurement in a 3D conformation structure involving two different cell types is complex, mainly performed manually, highly time consuming and examiner dependent. Standardization of encapsulated graft viability analysis before transplantation is a key point for the translation of the method from the bench side to clinical practice. In this study, we developed an automated analysis of islet viability based on confocal pictures processing of cells stained with three probes (Hoechst, propidium iodide, and PKH67). When compared with results obtained manually by different examiners, viability results show a high degree of similarity (under 3% of difference) and a tight correlation (r = 0.894; p < 0.001) between these two techniques. The automated technique offers the advantage of reducing the analysis time by 6 and avoids the examiner’s dependent variability factor. Thus, we developed a new efficient tool to standardize the analysis of islet viability in 3D structure involving several cell types, which is a key element for encapsulated graft analysis in clinical practice.
Islet transplantation is a promising therapy for millions of patients with type 1 diabetes as it offers a perspective of an efficient metabolic control with a prevention of severe hypoglycemia without insulin injections (1–3). However, islets graft success may be compromised by grafted islets exposition to an acute stress due to ischemia reperfusion injuries associated with an instant blood-mediated inflammatory reaction and to chronic rejection (4–9). Moreover, to prevent graft rejection, an immunosuppressive therapy is required, which can lead to severe adverse events such as infection or neoplasic disease.
To avoid immunosuppressant therapy and to maintain graft viability and functionality over time, several approaches are currently studied. Among them, islet encapsulation in a permeable three-dimensional structure seems highly promising (10). The capsule allows the diffusion of insulin, oxygen, and nutriments while protecting islets from the immune system (11). This technique has already shown interesting results in different species (12–14) and paves the way for free-immunosuppressive islets transplant therapy (15). To increase the efficacy of encapsulation, Veriter et al. have shown the benefits of co-encapsulating islets with mesenchymal stem cells (MSCs) (16). Indeed, MSCs are well known for their in vitro and in vivo protective effects on islets viability (17) and functionality (18) as well as their pro angiogenic effect (19) under different stress conditions (20). Thus, co-encapsulation of islets with MSCs appears to be a promising approach to improve treatment strategies for diabetic patients (21).
However, the translation from the bench side to patients is facing various obstacles (22). The validation of these complex capsules containing two different cell types requires an accurate analysis of the viability and functionality of the co-encapsulated cells. The reference method to evaluate islets viability is the fluorescein diacetate (FDA) and propidium iodide (PI) or ethidium bromide (EB) co-staining (23, 24). This method, usually performed by islet isolation centers to characterize and validate islet preparation before transplantation, can be limited in a complex multiple cell type culture system. Indeed, this double staining does not allow to discriminate the viability of the two different cells types. The other probes used to replace FDA as Calcium AM, SYTO®13, SYTO®24, and SYBR®14 are also limited to a monoculture system when coupled to IP/EB (25–27). These probes can be used for flow cytometry viability analysis, but the dissociation of the capsule is compulsory as well as the dissociation of islets before acquisition. This dissociation enables a precise analysis but could induce a cellular stress which could lead to altered viability analysis. Thus, the current manual counting under microscope of the stained cells is the only tool available to determine the viability of islets included in an unmodified 3D structure such as a capsule. This technique is operator dependant and highly time consuming (28) which is poorly adapted in the emergency context imposed by transplantation.
Due to these limitations, most of the current studies on islets co-encapsulation are mainly based on an evaluation of the insulin secretion functionality (26, 29) restricting analysis of the capsules to functionality analysis and neglecting cells viability analysis. Thus, the development of a new standardized tool to characterize the viability of each cell type in the capsule content is highly suitable and constitutes a key point to further develop this treatment strategy.
The aim of this study was to develop and validate a standardized method to substitute the current manual viability analysis of islets co-encapsulated with MSCs, in agreement with the clinical constraints. For that purpose, we propose an automated picture analysis of stained cells obtained by confocal microscopy.
Materials and Methods
Two adult Wistar male rats (350 g) were used in this study. Procedures were carried out in accordance with European Directives 86/609/EEC, 2010/63/UE well-being and treatment of animals. All the procedures were approved by the ethics committee affiliated to the animal facility of the university (D3842110001) and agreed by the French Ministry of Research (9998_LBFA-U1055).
Islets Isolation Procedure and Islets Culture
A cannula is inserted in the rat’s Wirsung duct to permit the pancreas perfusion with collagenase type IX, 1 mg/ml (Sigma). Perfused pancreas was removed and digested in warm bath at 37°C for 11 min as previously described (30). Islets were then purified by Histopaque (Sigma-Aldrich, St. Louis, MO, USA) density gradient centrifugation. Islets are seeded in RPMI 1640 supplemented by 10% fetal bovine serum (Sigma-Aldrich, St. Louis, MO, USA), 2 mM l-glutamine, 100 U/ml penicillin, 100 µg/ml streptomycin, and 1 mM sodium pyruvate (Pan Biotech, Aidenbach, Germany). Islets were incubated at 37°C in a humidified atmosphere (95% air–5% CO2) and cultured for 12–24 h before encapsulation.
Mesenchymal stem cells were isolated from bone marrow aspirate from four healthy donors who gave their written informed consent. All procedures were in compliance with the French public health code (Article L1241-1) and were performed by the Cell Therapy and Engineering Unit of EFS Auvergne Rhone Alpes. After thawing, MSCs are seeded at 3,500 cells/cm2. Cells are maintained in culture at 37°C and 5% CO2 in MEM alpha media (Sigma-Aldrich, St. Louis, MO, USA) supplemented by 10% fetal bovine serum (Sigma-Aldrich, St. Louis, MO, USA), 2 mM glutamax, 100 U/ml penicillin, and 100 µg/ml streptomycin (Thermo Fisher Scientific, Waltham, MA, USA). Media are changed every 3–4 days. When the cell density reached 80–90%, cells were split with Trypsin–EDTA (0.25%), phenol red (Thermo Fisher Scientific, Waltham, MA, USA).
All islets and MSCs are encapsulated in one session at the Laboratoire d’électronique des technologies de l’information of the Commissariat à l’Energie Atomique et aux énergies alternatives (CEA-LETI, Grenoble, France) according to a microfluidic method. The pancreatic islets are washed in a buffer combining 150 mM NaCl, 20 mM HEPES, and 1 µg glucose. The islets are included in a solution of 2.5% alginate, 150 mM NaCl, 20 mM HEPES, 1 g/l glucose, and encapsulated in the microfluidic encapsulation system. All the steps are integrated in a single-use cartridge allowing oil phase capsule formation, gelation of the capsules in a calcium bath, and harvesting in the buffer. The encapsulated islets are returned to culture in complete RPMI 1640 medium.
To differentiate the two cell types included in the capsule, MSCs were stained by PKH67 (Sigma-Aldrich, St. Louis, MO, USA) that labeled lipid membrane structure (λexc\λem = 490/504 nm) before encapsulation. Briefly, as previously described (31) MSCs were dissociated with trypsin 1× and wash with Dulbecco’s phosphate-buffered serum (PAA Laboratories, Velizy Villacoublay, France). Then cells were suspended in diluent C with 2 µM of PKH67 during 5 min. The staining reaction was stopped by adding fetal bovine serum. MSCs were then washed three times with complete media before encapsulation.
Before image acquisition, capsules containing cells were stained with 2.21 × 10−10 mol/ml Hoechst 33342 (λex/λem = 361/497 nm) during 15 min. Just before pictures’ acquisition, 1.50 × 10−10 mol/ml of PI (λex/λem = 545/630 nm) (Interchim, Montluçon, France) was added to the medium containing capsules. Hoechst stains nuclei of living cells in blue while PI stains nuclei of necrotic cells in red. MSCs previously stained by PKH67 are labeled in green (32). Thus, co-localization of PI or Hoechst stain along with the PKH67 signal permits to analyze MSCs’ viability whereas nuclei out of the green signal provides data regarding islets cells viability.
Images were collected with an LSM710 AxioObserver Airyscan inverted laser scanning confocal microscope (Carl Zeiss Enterprise, Oberkochen, Germany) equipped with a 20× oil dry objective (Plan apochromat 20×/0.8). Laser excitation was 405 nm for Hoechst, 488 nm for PKH67, and 561 nm for PI. Fluorescence emission detected by GaAsP detectors was 410–485 nm for Hoechst, 493–551 nm for PKH67, and 571–640 nm for PI. Confocal pinhole (Airy units) was 1 for all channels. Five z-stacks of 13–22 pictures (total = 77 pictures) with 10 µm z-step of optical sections were acquired and saved in TIFF format (1,024 × 1,024 pixels).
Manual Viability Analysis
To avoid bias, manual analysis were blinded and performed by four different examiners which did not know the results obtained with the automatized method. Each experimenter merged the green channel (corresponding to the MSC cytosols) with the blue or red channel. Then, selection of nuclei present in or out the green signal was visually performed on each picture of the stacks using the “Multi-point” tool of ImageJ 1.51. ImageJ is an open source software especially developed for picture analysis and widely used in the world (33). The viabilities presented in this article are calculated by dividing the number of nuclei stained with Hoechst by the total number of nuclei (number of nuclei stained with PI + Hoechst).
Automated Viability Analysis
Regarding our automated method developed, previously acquired pictures were analyzed by a code elaborated with the ImageJ interface. The description of the code can be separated into four main steps as presented in Figure 1.
Figure 1. Graphic representation of the automated analyses of the green ① and blue channel ② to detect living nuclei of islets and mesenchymal stem cells.
Once the software is started, a window is displayed to set up several variables. First, fields to fill in allow to entry the folder’s pathway containing the picture files, name, and numeration. This window also permits the configuration of the channel appellation and different filters concerning the minimal and maximal thresholds accepted during the binary transformation, the factor of regions of interest (ROIs) dilatation containing the detected MSCs, and a filter to suppress signal under the determined surface. Two check boxes allow to enhance the blue channel contrast if the signal is too low and to choose or not to analyze the viability of MSCs.
Then, the code loads the green channel, corresponding to the MSCs cytosol staining. The picture is transformed into a binary signal; the detection of the green areas is performed and saved as ROI. All ROIs detected are saved in the folder named “…/Analyse/ROI_Save.” Then, blue channel, corresponding to the living nuclei is loaded and transformed into a binary picture. In each ROI previously saved, the x, y coordinates of MSCs’ nuclei are detected and saved in a matrix. The same analysis is performed with the red channel. The next step corresponds to the suppression of the MSCs’ nuclei in blue and the red channel. For that step, we used the “MagicWand” function allowing a selection of all the black dots around the different x, y coordinates and replacing them by white dots. To avoid artifacts due to the analysis of the same nucleus on several slides, the code is able to detect if nuclei are present on the next slide at the exact location of those previously detected and if so, to delete the signal. Then, each picture of blue channel is saved in “…/Analyse/Islets_Nucleus” and red channel in “…/Analyse/Necrotic_Islets_Nucleus.” If the option “Analyse MSCs parameters:” is checked at the beginning of the program, the nuclei present in ROIs are analyzed and save in the folder “…/Analyse/MSC_Nucleus” for blue channel, and “…/Analyse/Necrotic_Islets_Nucleus” for the red channel. Finally, the number and area of the islets and MSCs’ nuclei detected in each picture are saved in the folder “…/Analyse.” An algorithmic representation of the software main steps and the entire code are available in open source, downloadable in Supplementary Material.
All data are presented in Figure 3A as mean ± SD and were analyzed using a one-way ANOVA as they were normally distributed. The correlation coefficient represented in Figure 3B was calculated as a Pearson product moment.
Figure 1 represents intermediate steps used by the software to analyze the different cell type’s viability. The Figure 1① shows the transformation of the initial green channel picture to a binary signal to allow the detection of the green areas. The last picture of Figure 1① shows the ROIs that are automatically saved by the software during analysis. In Figure 1②, the three mains steps of the blue channel analysis are as follows: the binary conversion, the selection of islets or MSCs’ nuclei, and the analysis are displayed. Once the blue channel is transformed in a binary signal, the picture is merged with the ROIs previously detected. Nuclei of MSCs or islets are automatically selected as black dots and black dots are segmented to precisely count the number of nuclei. Modified pictures used for analysis are those saved by the software.
Figure 2 corresponds to the three channels merged in an initial picture (Figure 2A) and the identification of each nucleus by the software (Figure 2B). Nuclei considered by the software as living islets appear in blue whereas MSCs appear in purple. Islets nuclei detected as necrotic appear in red whereas MSCs are drawn in orange. This figure allows to visually confirm the automated identification performed by the software.
Figure 2. Example of a picture obtained by confocal microscopy (A) and the identification of the different nuclei detected by the software (B). Red: necrotic nuclei (IP); green: mesenchymal stem cells cytosol (Pkh); and blue: living nuclei (Hoechst stain).
In Figure 3A, the mean viability of islets and MSCs detected manually is compared with the automated software approach using five stacks of pictures containing between 14 and 22 optical sections. This figure shows no difference between the viability of the islets detected manually or detected by the software (76.02 ± 6.3 vs. 77.07 ± 10.6%, respectively, ns). No differences were observed when comparing MSCs’ viability analyzed with the two techniques (manually: 83.20 ± 6.5% vs. software: 80.74 ± 8.8%, ns). The visual resemblance between these two techniques, showed in Figures 2 and 3A, is confirmed by the tight correlation presented in Figure 3B. Indeed, the correlation of the viability manually evaluated with the viability automatically determined by the software for MSCs and islets shows a coefficient of correlation over 0.89 and a p-value lower than 0.001.
Figure 3. Comparison of the islets (black) and MSCs (white) viability obtained manually or with the software (A). Correlation between these two methods for islets (dots) and MSCs (diamonds) cells (B). Errors bars of panel (B) correspond to variability of viabilities manually obtained by different experimenters. Results represent the mean ± SD; n = 5. Abbreviation: MSCs: mesenchymal stem cells.
This study provides evidence of the efficacy and reliability of the automatized techniques we have developed using the ImageJ interface to analysis cells viability in a 3D structure. The high similarity of viability measured by two different methods (less than 3% of variation) and the tight correlation between them (r = 0.894; p < 0.001) validate the results obtained with our technique. Moreover, the automated analysis tool suppresses the variability of the viability measured by different examiners (more than 5% in this study, see SD in Figure 3B). The standardization of viability analysis is a key point to determine common standardized quality criteria for islet preparation.
The automated method developed in this study allows to strongly reduce the time spent for analysis in comparison with the manual technique. Our results showed a decrease of the analysis duration from approximately 1 h 45 min to 15 min for users trained to the macro’s utilization. The saving process during the analysis steps allows a visual feedback of the software results, and a check of analysis if desired.
The availability of the code in Supplemental Material allows users with programming skills to perform few modifications to adapt the software to many different applications. Thus, this code allows the viability measurement of different cells type in all complex tissues which cannot be mechanically separated.
As co-encapsulation of islets and MSCs appears to be a highly promising technique to major the efficacy of islets graft in the diabetes medical care, we have developed a new standardized approach to measure and distinguish the viability of co-encapsulated cells through an automated and time saving method preserving the 3D conformation of the structure.
This study was carried out in accordance with the recommendations of European Directives 86/609/EEC, 2010/63/UE well-being and treatment of animals. The protocol was approved by the “Comité d’éthique en expérimentation animale de Grenoble” and by the French Ministry of Research (No. 9998_LBFA-U1055).
CC and CL participated equally in the design, measurement, analysis, and redaction of this work. AF, CC-R, and MO-P contributed to the data measurement and analysis. FR, AM, EF, and P-YB participated to the design of this work. SL and ET participated to the design and the redaction of this work.
Conflict of Interest Statement
The authors declare that the research was conducted in the absence of any commercial or financial relationships that could be construed as a potential conflict of interest.
CL was supported by a 3-year Ph.D. fellowship by the foundation “Agir Pour les Maladies Chroniques,” Grenoble, France. The authors thank Yasmina Saoudi and the Photonic Imaging Center of Grenoble Institute of Neuroscience PIC-GIN for the images acquisition and Maxime Pierron from Microsyst. for Biol. & Health Department—CEA-LETI—for cells encapsulation. The authors also thank Cindy Tellier and Pauline Durrenbach for the animal care during this study.
The Supplementary Material for this article can be found online at https://www.frontiersin.org/articles/10.3389/fendo.2018.00272/full#supplementary-material.
1. Tharavanij T, Betancourt A, Messinger S, Cure P, Leitao CB, Baidal DA, et al. Improved long-term health-related quality of life after islet transplantation. Transplantation (2008) 86:1161–7. doi:10.1097/TP.0b013e31818a7f45
2. Lablanche S, Borot S, Wojtusciszyn A, Bayle F, Tetaz R, Badet L, et al. Five-year metabolic, functional, and safety results of patients with type 1 diabetes transplanted with allogenic islets within the Swiss-French GRAGIL Network. diabetes Care (2015) 38:1714–22. doi:10.2337/dc15-0094
3. Barton FB, Rickels MR, Alejandro R, Hering BJ, Wease S, Naziruddin B, et al. Improvement in outcomes of clinical islet transplantation: 1999-2010. Diabetes Care (2012) 35:1436–45. doi:10.2337/dc12-0063
4. Lablanche S, Cottet-Rousselle C, Argaud L, Laporte C, Lamarche F, Richard MJ, et al. Respective effects of oxygen and energy substrate deprivation on beta cell viability. Biochim Biophys Acta (2015) 1847:629–39. doi:10.1016/j.bbabio.2015.04.002
5. Montolio M, Biarnes M, Tellez N, Escoriza J, Soler J, Montanya E. Interleukin-1beta and inducible form of nitric oxide synthase expression in early syngeneic islet transplantation. J Endocrinol (2007) 192:169–77. doi:10.1677/joe.1.06968
8. Shapiro AM, Ricordi C, Hering BJ, Auchincloss H, Lindblad R, Robertson RP, et al. International trial of the Edmonton protocol for islet transplantation. N Engl J Med (2006) 355:1318–30. doi:10.1056/NEJMoa061267
9. Moberg L, Johansson H, Lukinius A, Berne C, Foss A, Kallen R, et al. Production of tissue factor by pancreatic islet cells as a trigger of detrimental thrombotic reactions in clinical islet transplantation. Lancet (2002) 360:2039–45. doi:10.1016/S0140-6736(02)12020-4
12. Rengifo HR, Giraldo JA, Labrada I, Stabler CL. Long-term survival of allograft murine islets coated via covalently stabilized polymers. Adv Healthc Mater (2014) 3:1061–70. doi:10.1002/adhm.201300573
14. Haque MR, Kim J, Park H, Lee HS, Lee KW, Al-Hilal TA, et al. Xenotransplantation of layer-by-layer encapsulated non-human primate islets with a specified immunosuppressive drug protocol. J Control Release (2017) 258:10–21. doi:10.1016/j.jconrel.2017.04.021
15. Dufrane D, Goebbels RM, Gianello P. Alginate macroencapsulation of pig islets allows correction of streptozotocin-induced diabetes in primates up to 6 months without immunosuppression. Transplantation (2010) 90:1054–62. doi:10.1097/TP.0b013e3181f6e267
16. Veriter S, Gianello P, Igarashi Y, Beaurin G, Ghyselinck A, Aouassar N, et al. Improvement of subcutaneous bioartificial pancreas vascularization and function by coencapsulation of pig islets and mesenchymal stem cells in primates. Cell Transplant (2014) 23:1349–64. doi:10.3727/096368913X663550
18. Bal T, Nazli C, Okcu A, Duruksu G, Karaoz E, Kizilel S. Mesenchymal stem cells and ligand incorporation in biomimetic poly(ethylene glycol) hydrogels significantly improve insulin secretion from pancreatic islets. J Tissue Eng Regen Med (2017) 11:694–703. doi:10.1002/term.1965
19. Johansson U, Rasmusson I, Niclou SP, Forslund N, Gustavsson L, Nilsson B, et al. Formation of composite endothelial cell-mesenchymal stem cell islets: a novel approach to promote islet revascularization. Diabetes (2008) 57:2393–401. doi:10.2337/db07-0981
20. Yeung TY, Seeberger KL, Kin T, Adesida A, Jomha N, Shapiro AM, et al. Human mesenchymal stem cells protect human islets from pro-inflammatory cytokines. PLoS One (2012) 7:e38189. doi:10.1371/journal.pone.0038189
21. Hirabaru M, Kuroki T, Adachi T, Kitasato A, Ono S, Tanaka T, et al. A method for performing islet transplantation using tissue-engineered sheets of islets and mesenchymal stem cells. Tissue Eng Part C Methods (2015) 21:1205–15. doi:10.1089/ten.tec.2015.0035
22. Krishnan R, Ko D, Foster CE III, Liu W, Smink AM, de Haan B, et al. Immunological challenges facing translation of alginate encapsulated porcine islet xenotransplantation to human clinical trials. Methods Mol Biol (2017) 1479:305–33. doi:10.1007/978-1-4939-6364-5_24
23. Miyamoto M, Morimoto Y, Nozawa Y, Balamurugan AN, Xu B, Inoue K. Establishment of fluorescein diacetate and ethidium bromide (FDAEB) assay for quality assessment of isolated islets. Cell Transplant (2000) 9:681–6. doi:10.1177/096368970000900514
25. Boyd V, Cholewa OM, Papas KK. Limitations in the use of fluorescein diacetate/propidium iodide (FDA/PI) and cell permeable nucleic acid stains for viability measurements of isolated islets of langerhans. Curr Trends Biotechnol Pharm (2008) 2:66–84.
26. Davis NE, Beenken-Rothkopf LN, Mirsoian A, Kojic N, Kaplan DL, Barron AE, et al. Enhanced function of pancreatic islets co-encapsulated with ECM proteins and mesenchymal stromal cells in a silk hydrogel. Biomaterials (2012) 33:6691–7. doi:10.1016/j.biomaterials.2012.06.015
27. Bhaiji T, Zhi ZL, Pickup JC. Improving cellular function and immune protection via layer-by-layer nanocoating of pancreatic islet beta-cell spheroids cocultured with mesenchymal stem cells. J Biomed Mater Res A (2012) 100:1628–36. doi:10.1002/jbm.a.34111
28. Karaoz E, Genc ZS, Demircan PC, Aksoy A, Duruksu G. Protection of rat pancreatic islet function and viability by coculture with rat bone marrow-derived mesenchymal stem cells. Cell Death Dis (2010) 1:e36. doi:10.1038/cddis.2010.14
29. Yoshimatsu G, Sakata N, Tsuchiya H, Ishida M, Motoi F, Egawa S, et al. Development of polyvinyl alcohol bioartificial pancreas with rat islets and mesenchymal stem cells. Transplant Proc (2013) 45:1875–80. doi:10.1016/j.transproceed.2013.01.043
30. Muller YD, Mai G, Morel P, Serre-Beinier V, Gonelle-Gispert C, Yung GP, et al. Anti-CD154 mAb and rapamycin induce T regulatory cell mediated tolerance in rat-to-mouse islet transplantation. PLoS One (2010) 5:e10352. doi:10.1371/journal.pone.0010352
31. Rousselle C, Barbier M, Comte VV, Alcouffe C, Clement-Lacroix J, Chancel G, et al. Innocuousness and intracellular distribution of PKH67: a fluorescent probe for cell proliferation assessment. In Vitro Cell Dev Biol Anim (2001) 37:646–55. doi:10.1290/1071-2690(2001)037<0646:IAIDOP>2.0.CO;2
32. Nagyova M, Slovinska L, Blasko J, Grulova I, Kuricova M, Cigankova V, et al. A comparative study of PKH67, DiI, and BrdU labeling techniques for tracing rat mesenchymal stem cells. In Vitro Cell Dev Biol Anim (2014) 50:656–63. doi:10.1007/s11626-014-9750-5
Keywords: pancreatic islets, transplantation, co-encapsulation, mesenchymal stem cells, automatization, viability analysis
Citation: Chabert C, Laporte C, Fertin A, Tubbs E, Cottet-Rousselle C, Rivera F, Orhant-Prioux M, Moisan A, Fontaine E, Benhamou P-Y and Lablanche S (2018) New Automatized Method of 3D Multiculture Viability Analysis Based on Confocal Imagery: Application to Islets and Mesenchymal Stem Cells Co-Encapsulation. Front. Endocrinol. 9:272. doi: 10.3389/fendo.2018.00272
Received: 27 February 2018; Accepted: 09 May 2018;
Published: 25 May 2018
Edited by:Åke Sjöholm, Gävle Hospital, Sweden
Reviewed by:Guido Moll, Charité Universitätsmedizin Berlin, Germany
Hiroki Mizukami, Hirosaki University, Japan
Elisabetta Ferretti, Sapienza University, Italy
Copyright: © 2018 Chabert, Laporte, Fertin, Tubbs, Cottet-Rousselle, Rivera, Orhant-Prioux, Moisan, Fontaine, Benhamou and Lablanche. This is an open-access article distributed under the terms of the Creative Commons Attribution License (CC BY). The use, distribution or reproduction in other forums is permitted, provided the original author(s) and the copyright owner are credited and that the original publication in this journal is cited, in accordance with accepted academic practice. No use, distribution or reproduction is permitted which does not comply with these terms.
*Correspondence: Clovis Chabert, email@example.com
†These authors have contributed equally to this work.