Original Research ARTICLE
Evaluation of Rapid Library Preparation Protocols for Whole Genome Sequencing Based Outbreak Investigation
- 1Division of Clinical Bacteriology and Mycology, University Hospital Basel, Basel, Switzerland
- 2Applied Microbiology Research, Department of Biomedicine, University of Basel, Basel, Switzerland
- 3DBM Bioinformatics Core Facility, SIB Swiss Institute of Bioinformatics, Basel, Switzerland
- 4Personalized Health Basel, University of Basel, Basel, Switzerland
Whole genome sequencing (WGS) has become the new gold standard for bacterial outbreak investigation, due to the high resolution available for typing. While sequencing is currently predominantly performed on Illumina devices, the preceding library preparation can be performed using various protocols. Enzymatic fragmentation library preparation protocols are fast, have minimal hands-on time, and work with small quantities of DNA. The aim of our study was to compare three library preparation protocols for molecular typing: Nextera XT (Illumina); Nextera Flex (Illumina); and QIAseq FX (Qiagen). We selected 12 ATCC strains from human Gram-positive and Gram-negative pathogens with %G+C-content ranging from 27% (Fusobacterium nucleatum) to 73% (Micrococcus luteus), each having a high quality complete genome assembly available, to allow in-depth analysis of the resulting Illumina sequence data quality. Additionally, we selected isolates from previously analyzed cases of vancomycin-resistant Enterococcus faecium (VRE) (n = 7) and a local outbreak of Klebsiella aerogenes (n = 5). The number of protocol steps and time required were compared, in order to test the suitability for routine laboratory work. Data analyses were performed with standard tools commonly used in outbreak situations: Ridom SeqSphere+ for cgMLST; CLC genomics workbench for SNP analysis; and open source programs. Nextera Flex and QIAseq FX were found to be less sensitive than Nextera XT to variable %G+C-content, resulting in an almost uniform distribution of read-depth. Therefore, low coverage regions are reduced to a minimum resulting in a more complete representation of the genome. Thus, with these two protocols, more alleles were detected in the cgMLST analysis, producing a higher resolution of closely related isolates. Furthermore, they result in a more complete representation of accessory genes. In particular, the high data quality and relative simplicity of the workflow of Nextera Flex stood out in this comparison. This thorough comparison within an ISO/IEC 17025 accredited environment will be of interest to those aiming to optimize their clinical microbiological genome sequencing.
Whole genome sequences currently provide the highest resolution for typing bacterial pathogens. The implementation of next generation sequencing (NGS) in routine clinical microbiology laboratories provides the foundation to analyze bacteria with high resolution, reproducibility and accuracy. Decreasing costs and increasing ease of implementation through increasingly flexible platform options, means that more laboratories will seek this technology over time.
Whole genome sequencing (WGS) has shown its value in molecular epidemiology, from seminal papers on MRSA and Mycobacterium tuberculosis helping to trace and resolve epidemics (1, 2), to implementation in routine laboratories (3–5), and local molecular epidemiological studies (6, 7). Methods of analysis range from determination of multi-locus sequence type (MLST; low resolution) through core genome MLST (cgMLST; high resolution) to whole genome phylogenies based on single nucleotide polymorphisms (SNPs; highest resolution). Using WGS in outbreak detection ideally takes account of all mutations and genomic variability in order to fully resolve outbreak scenarios and transmission chains (5, 8–11). Factors encoded within the genomes, such as antimicrobial resistance (AMR) and virulence factors, can also be determined from good quality assemblies (3–5, 12). Quality assurance, backward compatibility, communication between experts in different fields, and reporting to clinicians are issues currently being addressed (13–17).
Behind all these analyzes lies the all-important data. Several technologies have been used over the past decade for WGS: Ion Torrent PGM, Roche 454, PacBio and most recently Oxford Nanopore Technologies. But it is predominantly data from Illumina machines, from the MiniSeq, MiSeq, NextSeq, or HiSeq platforms, that is used for molecular epidemiology or bacterial genomics, as evidenced by the vast amounts of Illumina data deposited in databases (>90% at the Short Read Archive). Prior to the sequencing step, DNA libraries need to be made, protocols for which can vary greatly. Given the relatively high cost of library preparation compared to sequencing, and the time required to perform it, library preparation is a critical and rate-limiting step. Although many aspects of WGS can be optimized for routine diagnostic microbiology (17), to date few studies have addressed the data quality produced by different library methods.
Mechanical shearing of DNA often offers the most even and controllable DNA fragmentation (18), but requires high amounts of input DNA and hands-on time. Automation of mechanical shearing is problematic, limiting throughput. The most popular and implementable library protocols use proprietary transposases to cleave the DNA and ligate the adapters in one step, a method which is rapid but dependent on the DNA/enzyme concentration ratio, and is subject to sequence bias. The impact of this bias on the %G+C rich Mycobacterium tuberculosis genome has been explored, and the TruSeq (Illumina) method, involving mechanical shearing of DNA, was found to be superior to the enzymatic Nextera XT (Illumina) (19). On the AT-rich Plasmodium falciparum genome, Nextera was again found to give highly biased results (20). This phenomenon has also been observed in human leukocyte antigen (HLA) genotyping (21).
With QIAseq FX, Qiagen have recently released a library preparation protocol that is based on fully enzymatic fragmentation (nuclease). The advantage of this approach is that the efficiency of the fragmentation is not as strongly affected by %G+C-content as the transposase from the Nextera XT approach. As QIAseq FX uses only an enzyme and not a whole complex, the adaptor ligation must then be applied in a separate step (QIAseq FX DNA Library Handbook). Another recent launch, Nextera Flex (Illumina) is also a transposome based library preparation kit, promising consistent yield and fragment size, and less sequence bias (22). The development over Nextera XT involves bead-conjugated transposomes, meaning that the tagmentation sites are positionally better defined by the DNA binding to the beads.
The costs of the different compared kits are quite similar, and up-to-date prices are listed on the manufacturer's websites. Currently, the difference across all is <20%. Some laboratories implement protocols using lower reagent volumes to reduce the per sample costs, however this study used the manufacturer's standard protocols.
Our aim was to compare the data quality from three commercial library preparation kits, for use in clinical routine microbiology WGS. The optimal protocol is rapid, performs consistently across all genome types without optimization, and produces high quality data for both rapid and reliable outbreak analysis and AMR gene detection.
Materials and Methods
In order to evaluate the usability of the different library preparation kits, we made a selection of 12 ATCC strains representing Gram-positive and negative pathogenic bacterial species, with a high range of %G+C-content (Table 1). A complete high-quality reference genome exists for each strain. Additionally, we included seven local patient isolates of Enterococcus faecium and five isolates from a Klebsiella aerogenes outbreak from 2018.
DNA Extraction and Sequencing
All work was performed in an ISO/IEC 17025 accredited environment, although only the Nextera XT protocol is currently accredited. DNA from all isolates was extracted by Qiagen EZ1 (Qiagen, Hilden, Germany) using the DNeasy blood and tissue kit (Qiagen), from a single colony. Prior to this, some isolate were subject to pretreatment: Mycobacterium tuberculosis was inactivated at 95°C for 1 h and disrupted in a TissueLyser (Qiagen) for 2 min at highest frequency; Streptococcus pyogenes was pre-treated with the TissueLyser (Qiagen) for 2 min at frequency 30; Staphylococcus were pre-treated with lysozyme und lysostaphin for 30 min at 37°C; all other bacteria were pre-treated using Proteinase K for 10 min at 56°C. Extracts were quantified by Qubit (Invitrogen), separated into three aliquots, and frozen at 20°C.
Libraries were created from the aliquots using Nextera XT (“XT”; Illumina), Nextera DNA Flex (“flex”; Illumina) or QIAseq FX (“Qia”; Qiagen). The recommended amounts and concentrations of DNA for each protocol were used where possible (1 ng for XT, 100 ng for Flex, 200 ng for Qia). To simulate a more realistic situation for M. tuberculosis, for which DNA extraction is not trivial, we used less DNA for the three kits (1 ng for XT, 10 ng for flex, 10 ng for Qia).
Each pool of libraries was loaded and sequenced separately on a NextSeq 500 device (cluster densities: XT 202, flex 189, Qia 244 K/mm2) and were sequenced using 2 × 151 bp paired end reads, within the Division of Clinical Microbiology, University Hospital Basel. The data was demultiplexed using bcl2fastq (version v18.104.22.168; Illumina).
Genomic Data Quality Analysis
Reads were trimmed using trimmomatic (version 0.38) (23) using default parameters (ILLUMINACLIP:2:30:10 SLIDINGWINDOW:4:15 MINLEN:125), and randomly subsampled using seqtk (version 1.3-r106, -s100; https://github.com/lh3/seqtk) to provide mean 10, 20, 50, 100, and 200-fold coverage of the genomes.
Assemblies were produced by unicycler (v0.3.0b) (24), with assembly parameters derived using QUAST (version 5.0.2) (25). The annotation was performed using Prokka (version 1.13) (26). AMR genes were predicted by using ABRicate (version 0.8.10; https://github.com/tseemann/abricate) with the NCBI database (accession: PRJNA313047).
Reads from ATCC strains were mapped using BWA (version 0.7.17) (27) against the complete references with all replicons concatenated (Table 1). The read depth at the different positions was determined using pilon (version 1.23). The insert size was calculated from the sam files using an in-house python script (https://github.com/danielwuethrich87/collection/blob/master/scripts/parse_sam_for_insertsize.py). The base-composition at the difference positions within the reads was calculated using FastQC (version 0.11.5; https://www.bioinformatics.babraham.ac.uk/projects/fastqc/) on the mapped 10-fold subsampled reads from ATCC25586.
K-mer signatures of sub-sampled reads and corresponding reference genomes were computed with Sourmash v2.0.0 (28) using the suggested MinHash resolution (1000:1 compression ratio) and a k-mer size of 31. The k-mer signature of the subsampled assemblies was assessed with the Jaccard distance metric, which is calculated by asking how many k-mers are shared between two samples vs. how many k-mers in total are in the combined samples [(Sample1 ∩ Sample2)/(Sample1 ∪ Sample2)]. A Jaccard distance of 1 means the samples are identical; a Jaccard distance of 0 means the samples are completely different. Overlaps with reference genome were also calculated in terms of containment [(Sample1 ∩ Sample2)/Sample1].
The orthologous groups were determined using the stand-alone Roary pipeline v3.12.0 (29), which takes annotated assemblies in GFF3 format produced by Prokka as above. Principal component analysis (PCA) was performed on the output table of gene presence/absence and the coordinates of the first two principal components (weighted by the proportion of variance explained) were used to calculate the distance of each sample from the reference as a metric to determine the similarity in terms of gene content.
For outbreak isolate genomes, data was analyzed in Ridom SeqSphere+ v4.1.6 for Enterococcus faecium cgMLST (30), and Klebsiella aerogenes cgMLST using an ad-hoc scheme comprising 3282 target loci based on the KCTC2190 genome (NC_015663.1) and 41 additional genomes from NCBI. Additionally, MentaLiST (version 1.0.0) (31) was used to identify the cgMLST alleles from the Enterococcus faecium isolates.
CLC Genomics Workbench 10.1 was used to generate Single Nucleotide Polymorphism (SNP) phylogenies. Mapping was performed using default parameters, variant calling used the parameters: 10x min coverage, 10 min count and 70% min frequency. SNP trees used a neighbor joining method: minimum coverage 10, minimum coverage 10%, minimum z-score 1.96, multi-nucleotide variants included. The mapping reference for the Klebsiella aerogenes outbreak was that of KCTC2190, accession number CP002824.
The Enterococcus faecium data was also analyzed using snippy (version 4.3.6, –minfrac 0.8; https://github.com/tseemann/snippy) for SNP calling comparing to the Aus0004 as reference (accession number NC_017022.1) For the phylogenetic analysis, only the core genome SNPs were used. The phylogenetic tree was calculated using the neighbor joining tree algorithm of the scikit-bio (version 0.2.0) package (http://scikit-bio.org/).
Library Preparation and Ease of Use in Routine Laboratories
We selected three different rapid library preparation kits, all of which are based on enzymatic fragmentation: Nextera XT (“XT”), Nextera DNA Flex (“flex”), and QIAseq FX DNA (“Qia”) as they each provide a complete solution kit. The required DNA input amount of the three kits is very variable: XT needs exactly 1 ng of input DNA; Qia and flex support a wide range of DNA inputs that can affect the library preparation. The insert size of the Qia kit can be controlled by adjusting the fragmentation time and DNA input amount (1–1,000 ng). Flex accepts a wide range of input DNA (1–500 ng) resulting in the same insert size (300–350 bp). However, in Qia and flex, different DNA input amounts require the number of cycles in the PCR amplification step to be adjusted. Qia also supports a PCR free protocol if more than 100 ng are applied. We decided to use 1 ng of input DNA for XT, 100 ng for flex and 200 ng for Qia. This amount of DNA is reliably produced by our routine DNA extraction techniques, and simplifies the Qia protocol through elimination of PCR. For M. tuberculosis we used only 10 ng for Qia and flex. For the Qia protocol we were aiming for a fragment peak size of 550 bp by using 6 min fragmentation time for 100 ng and 10 min for the 10 ng input. The other kits do not allow specific adjustments for fragment length in the standard protocol. Each of the final libraries using XT and Qia were quantified and the 24 samples were equimolarly pooled. As sample normalization is already included in flex, we pooled the samples by taking the same volume from each library.
The application of the three kits revealed their strengths and weaknesses in the laboratory. For routine work, time is of course a major factor. The provider of all three kits state that the library preparation takes 2.5 h. However, we were only able to reach this time with XT, and only if time taken for DNA quantification before and after is not included. We also have to mention that the XT protocol has been established in our laboratory for 3 years and therefore the technicians are highly experienced. The Qia and flex protocols both took ~4 h. The Qia kit requires long fragmentation time (~60 min) and ligation time (45 min). It also has to be considered that if the input DNA amount of Qia is below 100 ng, PCR and clean-up must be included, which adds a further 90 min. For flex, the resuspension of the beads with the transposomes requires optimization, as they stick to the walls of PCR plates. Saving hands-on time, especially with larger sample numbers, flex includes bead-based concentration normalization. Also of importance in routine laboratory work, the Illumina kits provide plenty of consumable, which allows for potential inaccurate pipetting and still allows the indicated number of samples to be processed. In contrast, the Qia fragmentation mixture volume delivered in the kit was too limited and resulted in the sequencing failure of one sample (NMB004375).
Taken together, XT has the most convenient protocol to use in the laboratory. However, flex provides some features that allow a very streamlined process. Even though the flex protocol takes longer than XT, the wide range of DNA input amount and the normalized output can lead to a significant time gain. The Qia protocol take also longer than the XT protocol and needs more adjustments according to the DNA input amounts. On the other hand, it offers the easy adjustment of insert sizes.
Genome Coverage Evaluation of ATCC Strains
As a first estimate for the quality of the sequencing we mapped the reads of the ATCC strains against their published reference genomes. For this purpose, we aligned the 100-fold subsampled reads from each sample to the reference and visualized the read depth distribution (Figure 1A). The read depth is most variable using XT. This is especially obvious in genomes with low %G+C-content, resulting in many genomic regions with low coverage. Qia and flex, on the other hand, show a more even distribution of read depth in all samples and therefore provide a more complete representation of the genomes. The unevenness of coverage is less pronounced in genomes with G+C-content of 40% or more: these show similar pictures with XT, flex and Qia.
Figure 1. Quality assessment of WGS data. (A) The reads of the three library kits subsampled to 100-fold were mapped against the 12 reference genomes and the read depth called was measured. The colors indicate the different library preparation kits. The x-axis reflects the position along the genomes and the y-axis the read depth. (B) The insert size of the different libraries was calculated using the alignment of the paired-end reads to the reference. The boxplots represent the calculations from the different species, with the lowest %G+C-content on the left, and the highest on the right. In the boxplots the lower and upper hinges correspond to the first and third quartiles. The whiskers are located at 1.5x of the interquartile range. (C) The base composition of all the nucleotide sites in the reads was determined. The bases on the left side show the composition around the fragmentation site.
Based on the alignments of the reads to the reference genomes we calculated the insert sizes of the different library preparation kits (Figure 1B). With XT we see a clear trend that the genomes with higher %G+C-content have larger insert sizes, showing again that this method is highly sensitive to %G+C-content. The insert sizes of the flex and Qia are stable across the different genomes, with the exception of the low input Mycobacterium tuberculosis, and seem unaffected by %G+C-content. Using flex, the insert sizes are well above 300 bp, which allows an optimal use of 151 × 151 paired-end reads. With Qia we have an insert size slightly above 200 bp, despite having aimed for 400 bp (550 bp fragment size). This value should be able to be adjusted through in-depth protocol optimization.
Looking at %G+C-content variation within the reads (Figure 1C), overall the reads produced by Qia and flex are closer to the actual genomic %G+C-content than those from XT. Focusing on the beginning of the reads, which represent the fragmentation sites, flex and XT give a strong variation of the %G+C-content, which is characteristic for the transposome used by Nextera. Surprisingly this fragmentation preference does not affect the read depth distribution of flex. Using Qia we see that the beginning of the reads are very similar to the %G+C-content of the genomes.
Evaluation of Assembly Quality
In order to study the genome representation in the different library preparation kits, we analyzed the subsampled reads of the ATCC strains at mean 10, 20, 50, 100, and 200-fold coverage. K-mer containment was used to compare the k-mers in the reads of the difference subsamples against the k-mers in the reference assemblies (Figure 2A). With this analysis we found that, using Qia and flex with an average read depth of 10-fold, more than 99% of all k-mers were found in most of the genomes. At 50-fold with these two kits, k-mers were already completely saturated, indicating that all the genome is represented. XT shows a different picture: while increasing read depth increases the percentage of k-mers found, the k-mer pool of the reference is not completely represented using XT even with 100 and 200-fold coverage, indicating that there will always be regions absent, leading to incomplete representation of the genome.
Figure 2. Comparison of the sequencing content using k-mers. (A) All k-mers identified within the reads were compared to those k-mer from the reference genomes. The x-axis shows the different subsampling of the reads and the y-axis shows the percent of k-mers that were found in the reads. (B) The assemblies of the sequenced strains were compared against the reference assemblies using the Jaccard index of the k-mers. The x-axis shows the different subsampling of the reads used for each assembly. The y-axis shows the Jaccard index. The colors indicate the different library preparation kits. In the boxplots the lower and upper hinges correspond to the first and third quartiles. The whiskers are located at 1.5x of the interquartile range.
De novo assembly of the subsampled reads was performed, and the k-mers of the assemblies compared to those from the references using the Jaccard index (Figure 2B). This analysis shows a similar picture. Qia and flex show a good representation of the genome with a 50-fold coverage upwards, whereas using XT with 100-fold and 200-fold coverage, reads do not completely represent the genome.
Assembly quality measures (NG50, number of contigs, genome representation, mismatches) were calculated using Quast (Figure S1). With increasing coverage, contig length (NG50) increases, as does genome fraction compared against the reference genomes, the number of contigs in the assembly decreases, and so do the number of mismatches called between the assemblies and the references. This analysis again shows that we can obtain an almost complete representation of the genome with 50-fold coverage using Qia and flex; XT on the other hand needs 100-fold or more coverage. In order to compare the gene content of assemblies from the different library preparation kits and subsamples, we performed a PCA on the presence and absence of orthologous groups (Figure S2). In general, we found that low coverage assemblies (10- and 20-fold) are more likely to result in different gene content (less genes) to the references, which cluster with the high coverage assemblies. However, we also found that in strains with low %G+C-content (≤50.37%; ATCC25586, ATCC700819, ATCC25923, ATCC29212, ATCC19615, ATCC25845, and ATCC25922) the genes found in the XT assemblies, even at high coverage, are separated from the references, Qia and flex assemblies.
AMR genes were analyzed in the published complete ATCC reference genomes and the assemblies from our experiment, using ABRicate (Table 2). We found that we can find every resistance gene from flex and Qia reads if the coverage is 50-fold or over. With XT many genes are not found with a coverage of 50-fold and some genes are even absent from the assemblies produced from a coverage of 100 or 200-fold.
Estimation of Coverage Required for cgMLST Analysis
In 2018, we sequenced, as routine, several cases of vancomycin-resistant Enterococcus faecium (VRE) and a small outbreak of Klebsiella aerogenes (K. aerogenes) that was not associated with our hospital. For this study, we selected five K. aerogenes isolates and seven VRE isolates to evaluate the performance of the three kits on samples from the routine clinical microbiology laboratory.
After subsampling, we typed the seven VRE strains using the cgMLST scheme of Enterococcus faecium, in the commercial software Ridom SeqSphere+, and using the open source software MentaLiST. In order to determine the resolution, we compared the number of core genes found in each sample and subsample (Figures 3A,B). Using reads from Qia and flex libraries, most of the core genes are found with a 50-fold coverage and over. With XT, over 25% of the core genes are not identified using a coverage of 50-fold, and 10–20% are still missing at 100-fold coverage.
Figure 3. cgMLST alleles identified from the patient isolates. The different subsamples (x-axis) were used to determine of the alleles of the core genome. The different strains are depicted as bars. The y-axis shows the percentage of core genes that can be used for allelic typing. The colors indicate the different library preparation kits. The E. faecium isolates were analyzed using Mentalist (A) and Ridom SeqSphere+ (B). The K. aerogenes isolates were analyzed only using Ridom SeqSphere+ (C). The failed Qia library is labeled with “*”.
For the five K. aerogenes we created an ad-hoc cgMLST scheme using Ridom SeqSphere+. In comparison to the VRE, 50-fold coverage was sufficient for all three kits to assign alleles to over 85% of core genes (Figure 3C). As K. aerogenes has a higher %G+C-content than E. faecium, we have seen that this results in more equal genome coverage from all kits, especially XT, leading to better assemblies and increased core gene identification.
Analysis of Vancomycin-Resistant Enterococcus faecium Isolates
A previous investigation showed an outbreak of VRE from Switzerland carried the same MLST type (ST796) as an outbreak in Australia (32–34). Out of this investigation we selected four isolates from an outbreak, as well as three isolates (ST117) from an example of in-patient acquisition of a vancomycin-resistance carrying transposon (Tn1549) in the same strain background. To test the performance of the different kits we aligned the reads against a reference (Aus0004) for the construction of a SNP scheme. For the analysis we only selected SNPs from the core genome to reduce false SNPs caused by the distance to the reference. Using high coverage samples (≥50-fold Qia, 50-fold flex, 100-fold XT), we found more than 2,000 SNPs between the isolates of ST117 and ST796. In contrast, within each sequence type only 1-2 SNP differences were identified (Figure S3A). All strains showed the same distance to the reference at the root of the tree (Figure S3B). However, if we also include samples with lower sequencing depth (≤20-fold Qia, ≤20-fold flex, ≤50-fold XT), we find a higher diversity in the pairwise comparison of the strains (Figure S3C): up to 49 SNPs among ST117 isolates, and in up to 51 SNPs among the ST796 isolates. This is an indication that we are discovering falsely called SNPs. The neighbor joining phylogeny also shows that that subsamples with lower sequencing depth have a smaller distance to the root, as not all SNPs are called (Figure S3D). Therefore, we conclude that we can improve SNP typing: lowering the number of falsely called SNPs and increasing the number of “real” SNPs, by using higher sequencing coverage, and Qia and flex kits.
We identified AMR genes in the VRE isolates (Table S1). Using Qia and flex, all AMR genes are found if at least a coverage of 50-fold is used. For XT, a coverage of at least 100-fold is needed to ensure the detection of all genes.
Analysis of Klebsiella aerogenes Outbreak Analysis
Five K. aerogenes isolates from a small outbreak were investigated using the commercial software Ridom SeqSphere+ for cgMLST, and CLC genomics for SNP analysis. The cgMLST analysis gave the same results for the all library kits at 200-fold (Figure 4A), showing small numbers of allelic discriminations between the isolates. Using lower coverage subsampled datasets, the number of identified allelic differences becomes smaller (Figure S4). Using flex, all the strains could be differentiated even with 10-fold coverage, which was not the case for Qia and XT, where isolates began to collapse into clusters. In the SNP analysis we did not find any differences between the kits with 200-fold coverage (Figure 4B), with each identifying 15 SNPs separating the five isolates. However, if we perform the analysis with the lower subsampled reads, again the resolution declines (Figure 4C). We could still capture the whole diversity using flex with 100-fold and 50-fold coverage, and Qia with 100-fold. Using XT we identified 13 and 14 SNPs (1 SNP was falsely called) in the 100-fold and 50-fold dataset, respectively.
Figure 4. Analysis of the K. aerogenes outbreak isolates. (A) The isolates (200-fold subsamples) were analyzed using cgMLST in Ridom SeqSphere+ and are depicted in a minimum spanning tree (MST). The isolates are shown as circles. If two strains are identical they collapse into one circle. The numbers on the lines connecting the different circles show the number of different alleles between two isolates (not to scale). (B) The genomic distances between the isolates (200-fold subsamples) is show as a phylogenetic tree representing all SNP differences across the whole genome. (C) SNP numbers across the tree called using the different subsamples.
This in-depth comparison of three commercial library preparations kits shows the superiority of the Qia and flex kits over XT concerning the quality of the data produced. Through our strategic study design, including a range of human pathogens, we have shown that these two methods produce high quality NGS data that represent the whole genome, at a mean coverage of at least 50-fold. The fragmentation step of these methods is highly stable to variability in the %G+C-content in the genome, resulting in almost even distribution of read depth. In contrast, XT is highly affected by the %G+C-content variation within and between the genomes. This results in an incomplete representation of the genome, especially if lower read depths (<100-fold) are used. Therefore, we suggest that, while Qia and flex libraries can be relied on at mean coverages 50-fold and above, a higher sequencing depth for libraries prepared with XT is required (over 100-fold), which will affect the number of samples that can be pooled on a sequencing run (Table S2). This is crucial for the highest resolution of typing, and for comprehensive surveillance of genomic elements such as AMR and virulence genes. We note that we found limitations of the XT data in terms of genome representation in some cases even at a mean read depth of 200-fold.
Our study protocol used a single DNA extraction protocol, and as such we cannot exclude that the tested kits show a different performance with other protocols. Additionally, we conducted this study without technical replicates, therefore variability between batches could not be assessed.
The investigations of the E. faecium and K. aerogenes patient isolates show the strength of WGS for bacterial typing. Even though the three kits are based on significantly different enzymatic and chemical reactions, the typing results are identical between the methods at high coverage. If the data quality is low, resolution is lost, both in cgMLST and SNP analysis. This is a very important finding for typing laboratories, and especially large-scale projects that want to compile NGS data from nationwide labs to establish national surveillance (35). In all settings, however: local, national or global, the quality control and bioinformatic analysis remain key for epidemiological analysis, as low-quality data can affect the outcomes by lowering the resolution or allowing the false calling of SNPs.
Usability in the Laboratory
The evaluation of the usability in the laboratory showed that XT is the quickest protocol (2.5 h). The core protocols of Qia and flex take a least 1 h longer. However, in the flex protocol, library normalization is included, which reduces the time needed to pool the libraries. This protocol also offers a flexible input amount (50–500 ng) that does not require optimization, saving time in the DNA preparation. However, if <50 ng is available, the number of PCR cycles has to be increased. The Qia protocol needs an accurate measurement of the input DNA, as the resulting fragmentation depends on the DNA amount. If <100 ng input DNA is used, additional PCR and clean-up steps are required that prolong the library preparation by a further 60–90 min. Therefore, XT is superior in time efficiency, but closely followed by flex.
The fragment length is also a very important factor in the sequencing process. Libraries with insert sizes that are smaller than the read length lead to overlaps in the reads pair and therefore loss of sequence information. Fragment length also affects the cluster density calculation, the clustering efficiency and the sequencing depth. If the fragment length is stable across different sample types, the amount of DNA is sufficient data to calculate the molarity and therefore the number of clusters. Long fragments lead to inefficiency in the clustering and can result in very low cluster densities; short fragments can lead to over clustering and the failure of a run. Therefore, it is important to produce a stable insert size for the libraries, independent of the input DNA. Our comparisons showed that the insert length in Qia and flex are stable across varying %G+C-content. The insert size from the XT is much more affected by the %G+C-content. In our experience with XT, we obtain much higher cluster densities (occasionally leading to over clustering) when sequencing AT-rich species such as Campylobacter, as opposed to Klebsiella. It is worth mentioning that we suggest to use 2 × 150 bp reagent kits for these libraries, as the tested libraries generally show an insert size of <350 bp. We do not recommend using the MiSeq Reagent Kit v3 (600-cycle) for these libraries, as it produces 2 × 300 bp reads, and the resulting read pairs would overlap with libraries of this length.
Current trends indicate that WGS will be used more often in routine diagnostics and therefore also the number of samples processed will increase. Thus, an implementation of the library preparations kits on automated liquid handling systems will reduce the time and cost associated with this technology. All three protocols discussed in this study can be implemented on liquid handling systems that are equipped with a thermocycler and a magnetic stand.
We have summarized the important features of the different kits that should support other labs in deciding on the most appropriate library preparation kit (Table 3).
The evaluation of the three kits clearly showed that the data quality from libraries made with Qia and flex are superior to those from XT. The comparison of laboratory processes of the Qia and flex kits shows that flex is superior, as the protocol needs very few adjustments, and less hands-on time for routine questions. Therefore, flex best enables streamlining of the laboratory processes for WGS in the context of surveillance.
The datasets generated for this study can be found in the ENA repository under project PRJEB31421.
HS-S, FB, and DW performed the data analysis and wrote the manuscript. AC and JR performed the data analysis. AE wrote the manuscript.
FB was funded through a grant to AE from Personalized Health Basel.
Conflict of Interest Statement
The authors declare that the research was conducted in the absence of any commercial or financial relationships that could be construed as a potential conflict of interest.
We thank Christine Kiessling, Magdalena Schneider, Elisabeth Schultheiss, Clarisse Straub, and Rosa-Maria Vesco (University Hospital Basel) for excellent technical assistance.
The Supplementary Material for this article can be found online at: https://www.frontiersin.org/articles/10.3389/fpubh.2019.00241/full#supplementary-material
1. Gardy JL, Johnston JC, Ho Sui SJ, Cook VJ, Shah L, Brodkin E, et al. Whole-genome sequencing and social-network analysis of a tuberculosis outbreak. N Engl J Med. (2011) 364:730–9. doi: 10.1056/NEJMoa1003176
2. Harris SR, Cartwright EJ, Torok ME, Holden MT, Brown NM, Ogilvy-Stuart AL, et al. Whole-genome sequencing for analysis of an outbreak of meticillin-resistant Staphylococcus aureus: a descriptive study. Lancet Infect Dis. (2013) 13:130–6. doi: 10.1016/s1473-3099(12)70268-2
3. Deurenberg RH, Bathoorn E, Chlebowicz MA, Couto N, Ferdous M, Garcia-Cobos S, et al. Application of next generation sequencing in clinical microbiology and infection prevention. J Biotechnol. (2017) 243:16–24. doi: 10.1016/j.jbiotec.2016.12.022
4. Balloux F, Bronstad Brynildsrud O, van Dorp L, Shaw LP, Chen H, Harris KA, et al. From theory to practice: translating Whole-Genome Sequencing (WGS) into the clinic. Trends Microbiol. (2018) 26:1035–48. doi: 10.1016/j.tim.2018.08.004
5. Besser J, Carleton HA, Gerner-Smidt P, Lindsey RL, Trees E. Next-generation sequencing technologies and their application to the study and control of bacterial infections. Clin Microbiol Infect. (2018) 24:335–41. doi: 10.1016/j.cmi.2017.10.013
6. Hinic V, Seth-Smith H, Stockle M, Goldenberger D, Egli A. First report of sexually transmitted multi-drug resistant Shigella sonnei infections in Switzerland, investigated by whole genome sequencing. Swiss Med Wkly. (2018) 148:w14645. doi: 10.4414/smw.2018.14645
7. Wuthrich D, Gautsch S, Spieler-Denz R, Dubuis O, Gaia V, Moran-Gilad J, et al. Air-conditioner cooling towers as complex reservoirs and continuous source of Legionella pneumophila infection evidenced by a genomic analysis study in 2017, Switzerland. Euro Surveill. (2019) 24:1800192. doi: 10.2807/1560-7917.ES.2019.24.4.1800192.
9. Maiden MC, Jansen van Rensburg MJ, Bray JE, Earle SG, Ford SA, Jolley KA, et al. MLST revisited: the gene-by-gene approach to bacterial genomics. Nat Rev Microbiol. (2013) 11:728–36. doi: 10.1038/nrmicro3093
10. Pearce ME, Alikhan NF, Dallman TJ, Zhou Z, Grant K, Maiden MCJ. Comparative analysis of core genome MLST and SNP typing within a European Salmonella serovar Enteritidis outbreak. Int J Food Microbiol. (2018) 274:1–11. doi: 10.1016/j.ijfoodmicro.2018.02.023
11. Schurch AC, Arredondo-Alonso S, Willems RJL, Goering RV. Whole genome sequencing options for bacterial strain typing and epidemiologic analysis based on single nucleotide polymorphism versus gene-by-gene-based approaches. Clin Microbiol Infect. (2018) 24:350–4. doi: 10.1016/j.cmi.2017.12.016
13. Gargis AS, Kalman L, Lubin IM. Assuring the quality of next-generation sequencing in clinical microbiology and public health laboratories. J Clin Microbiol. (2016) 54:2857–65. doi: 10.1128/jcm.00949-16
15. Carrico JA, Rossi M, Moran-Gilad J, Van Domselaar G, Ramirez M. A primer on microbial bioinformatics for nonbioinformaticians. Clin Microbiol Infect. (2018) 24:342–9. doi: 10.1016/j.cmi.2017.12.015
16. Crisan A, McKee G, Munzner T, Gardy JL. Evidence-based design and evaluation of a whole genome sequencing clinical report for the reference microbiology laboratory. PeerJ. (2018) 6:e4218. doi: 10.7717/peerj.4218
17. Rossen JWA, Friedrich AW, Moran-Gilad J. Practical issues in implementing whole-genome-sequencing in routine diagnostic microbiology. Clin Microbiol Infect. (2018) 24:355–60. doi: 10.1016/j.cmi.2017.11.001
18. Quail MA, Kozarewa I, Smith F, Scally A, Stephens PJ, Durbin R, et al. A large genome center's improvements to the Illumina sequencing system. Nat Methods. (2008) 5:1005–10. doi: 10.1038/nmeth.1270
19. Tyler AD, Christianson S, Knox NC, Mabon P, Wolfe J, Van Domselaar G, et al. Comparison of sample preparation methods used for the next-generation sequencing of Mycobacterium tuberculosis. PLoS ONE. (2016) 11:e0148676. doi: 10.1371/journal.pone.0148676
20. Quail M, Smith M, Coupland P, Otto T, Harris S, Connor T, et al. A tale of three next generation sequencing platforms: comparison of Ion Torrent, Pacific Biosciences and Illumina MiSeq sequencers. BMC Genomics. (2012) 13:341. doi: 10.1186/1471-2164-13-341
21. Lan JH, Yin Y, Reed EF, Moua K, Thomas K, Zhang Q. Impact of three Illumina library construction methods on GC bias and HLA genotype calling. Hum Immunol. (2015) 76:166–75. doi: 10.1016/j.humimm.2014.12.016
22. Bruinsma S, Burgess J, Schlingman D, Czyz A, Morrell N, Ballenger C, et al. Bead-linked transposomes enable a normalization-free workflow for NGS library preparation. BMC Genomics. (2018) 19:722. doi: 10.1186/s12864-018-5096-9
24. Wick RR, Judd LM, Gorrie CL, Holt KE. Unicycler: resolving bacterial genome assemblies from short and long sequencing reads. PLoS Comput Biol. (2017) 13:e1005595. doi: 10.1371/journal.pcbi.1005595
30. de Been M, Pinholt M, Top J, Bletz S, Mellmann A, van Schaik W, et al. Core genome multilocus sequence typing scheme for high- resolution typing of Enterococcus faecium. J Clin Microbiol. (2015) 53:3788–97. doi: 10.1128/jcm.01946-15
32. Buultjens AH, Lam MM, Ballard S, Monk IR, Mahony AA, Grabsch EA, et al. Evolutionary origins of the emergent ST796 clone of vancomycin resistant Enterococcus faecium. PeerJ. (2017) 5:e2916. doi: 10.7717/peerj.2916
33. Mahony AA, Buultjens AH, Ballard SA, Grabsch EA, Xie S, Seemann T, et al. Vancomycin-resistant Enterococcus faecium sequence type 796 - rapid international dissemination of a new epidemic clone. Antimicrob Resist Infect Control. (2018) 7:44. doi: 10.1186/s13756-018-0335-z
34. Wassilew N, Seth-Smith HM, Rolli E, Fietze Y, Casanova C, Fuhrer U, et al. Outbreak of vancomycin-resistant Enterococcus faecium clone ST796, Switzerland, December 2017 to April 2018. Euro Surveill. (2018) 23:1800351. doi: 10.2807/1560-7917.Es.2018.23.29.1800351
35. Egli A, Blanc DS, Greub G, Keller PM, Lazarevic V, Lebrand A, et al. Improving the quality and workflow of bacterial genome sequencing and analysis: paving the way for a Switzerland-wide molecular epidemiological surveillance platform. Swiss Med Wkly. (2018) 148:w14693. doi: 10.4414/smw.2018.14693
Keywords: NGS, next generation sequencing, library, Illumina, whole genome sequencing, comparison, bacteria, prokaryotes
Citation: Seth-Smith HMB, Bonfiglio F, Cuénod A, Reist J, Egli A and Wüthrich D (2019) Evaluation of Rapid Library Preparation Protocols for Whole Genome Sequencing Based Outbreak Investigation. Front. Public Health 7:241. doi: 10.3389/fpubh.2019.00241
Received: 09 April 2019; Accepted: 12 August 2019;
Published: 27 August 2019.
Edited by:Vitali Sintchenko, University of Sydney, Australia
Reviewed by:Rebecca Rockett, University of Sydney, Australia
Avram Levy, University of Western Australia, Australia
Copyright © 2019 Seth-Smith, Bonfiglio, Cuénod, Reist, Egli and Wüthrich. This is an open-access article distributed under the terms of the Creative Commons Attribution License (CC BY). The use, distribution or reproduction in other forums is permitted, provided the original author(s) and the copyright owner(s) are credited and that the original publication in this journal is cited, in accordance with accepted academic practice. No use, distribution or reproduction is permitted which does not comply with these terms.
*Correspondence: Helena M. B. Seth-Smith, email@example.com